Serial number | Dynamic Range Detector slope | SFDR | Noise Figure | Sensitivity limit | Band pass filter | Allan time, Allan variance | Supply current | Location, user, comments and links |
---|---|---|---|---|---|---|---|---|

1st series of Callisto spectrometer based on Philips CD1316/IV. (HW manufactured by A. James, R. Moser, P. Behm, I. Zamora, S. Zihlmann, M. Nyirö, H. Temperli, M. Battaglia, A. Vydrzel, T. Moser and T. Kittelmann), 10.07.2009 - 31.05.2010 | ||||||||

eC00 (breadboard) | N/A | N/A | N/A | N/A | N/A | N/A | 225mA/12V | used for spare parts ... |

eC01 (prototype) at 24°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs time 607.19MHz (channel 130) full report | 222mA/12V | Refurbished 2020-06-11, sent to ASSA 2020-09-03 |

eC02 at 23.5°C | plot vs AGC-voltage | sfdr | NF | Sens | plot vs frequency | plot vs time 458.69MHz (channel 100) full report | 223mA/12V | San Jose, Costa Rica |

eC03 at 22.5°C | plot vs AGC-voltage | sfdr | NF | Sens | plot vs frequency | plot vs time 607.19MHz (channel 130) full report | 229mA/12V | University of Addis Ababa (AAU), Ethiopia |

eC04 at 22°C | plot at PWM=120 | sfdr | NF | Sens | plot vs frequency | N/A | 226mA/12V | HB9SCT, Freienbach |

eC05 at 22°C | plot at PWM=120 | sfdr | NF | Sens | plot vs frequency | N/A | 230mA/12V | Bleien 7m dish. |

eC06 at 22°C | plot at PWM=120 | sfdr | NF | Sens | plot vs frequency | N/A | 242mA/12V | UNAM, Mexico city |

eC07 at 22°C | plot at diff. PWM | sfdr | NF | Sens | plot vs frequency | plot vs time 607.19MHz | 226mA/12V | KASI Daejeon South Korea |

eC08 at 21°C | plot at diff. PWM | sfdr | NF | N/A | plot vs frequency | N/A | 220mA/12V | KASI South Korea (kit) |

eC09 at 21°C | plot at diff. PWM | sfdr | NF | Sens | plot vs frequency | plot vs time 800MHz | 216mA/12V | Royal Observatory of Belgium ROB |

eC10 at 21°C | plot at diff PWM | sfdr | NF | Sens | plot vs frequency | plot vs time 800MHz | 218mA/12V | Ooty India (U) |

eC11 at 21°C | plot at diff PWM | sfdr | NF | Sens | plot vs frequency | plot vs time 800MHz | 217mA/12V | Ooty India (V) |

eC12 at 21°C | plot at diff PWM | sfdr | NF | Sens | plot vs frequency | plot vs time 800MHz | 215mA/12V | Instrument refurbished hardware and firmware, now eC12x sent to Asunción + LNA2000/2, Paraguay 2021-07-26. |

eC13 at 21°C | plot at diff PWM | sfdr | NF | Sens | plot vs frequency | plot vs time 800MHz | 214mA/12V | OSRA, Ondrejov, Czech Republic |

eC14 at 21°C | plot at diff PWM | sfdr | NF | Sens | plot vs frequency | plot vs time 800MHz | 215mA/12V | Royal Observatory of Belgium ROB [Rep.] (Data) |

eC15 at 21°C | plot at diff PWM | sfdr | NF | Sens | plot vs frequency | plot vs time 800MHz | 213mA/12V | UoM Mauritius main instrument MRT2 |

eC15a at 20°C | plot at PWM=80 | sfdr | NF | Sens | plot vs frequency | nil | 236mA/12V | Repair and upgrade 2017-04-26 (all IC, Quartz 25.43 MHz, R3, R4) Repair 2019-04-09 (all digital IC replaced) |

eC16 at 21°C | plot at diff PWM | sfdr | NF | Sens | plot vs frequency | plot vs time 800MHz | 213mA/12V | Gauribidanur India (replacement for eC01) |

eC17 at 21°C | plot at diff PWM | sfdr | NF | Sens | plot vs frequency | plot vs time 800MHz | 214mA/12V 213mA/12V |
RCAG Ulaan Baatar Mongolia (1st instrument) Test 09.08.2017, everything o.k. |

eC18 at 21°C | plot at diff PWM | sfdr | NF | Sens | plot vs frequency | plot vs time 800MHz | 212mA/12V | HAM-Version, BW = 2.4 MHz, HB9SCT, Freienbach Switzerland. |

eC19 at 19°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 233mA/12V | TCD Dublin, Ireland (Blog) |

eC20 at 20°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 219mA/12V | INPE São José dos Campos, Brasil |

eC21 at 18°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 222mA/12V | New ATmega16 V1.8, new EEPROM, + 2 Heat sinks 7805 22.08.2019 Installed with LWA in Landschlacht |

eC22 at 20°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 212mA/12V | Refurbished 2018-12-05: new ATmega16, new EEPROM, new quartz, new DC-connector Reserved for POLAND (Gürkan Myczko) with up-converter 5-55->105-155 MHz, G5RV |

eC23 | N/A | N/A | N/A | N/A | N/A | N/A | N/A | Kit for Melbourne, Australia (Mihai Dumitru), now at LMRO |

eC24 | N/A | N/A | N/A | N/A | N/A | N/A | N/A | Kit for Perth, Australia (Hans Michelmayr), 2020-09-03 checked and sent back to ASSA with new crystal, ATmega16 V1.8 on 03-09-2020. |

eC25 at 23°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 222mA/12V | Monitoring 5m dish Bleien. Upgrade to V1.8, new ATmega16, new EEPROM, new quartz 25.43 MHz and 2 heat sinks 2021-06-30. |

eC26 at 23°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 222mA/12V | New ATmega16 V1.8, new EEPROM, + 2 Heat sinks 7805 22.08.2019 Installed with LWA in Landschlacht. |

eC26x at 27°C | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 242mA/12V | Lost and found in the lab on a demonstration frame. Refurbished with new enclosure. Sent to ASSA 2020-09-03. |

eC27 at 23°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 220mA/12V | ERAC Heidelberg, Germany |

eC28 at 23°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 222mA/12V | INPE São José dos Campos, Brasil |

eC29 at 23°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 220mA/12V | Jim Sky, Hawaii |

eC30 at 22°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 215mA/12V | DL2RD Berlin -> ANGKASA Malaysia |

eC31 at 22°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 215mA/12V | Helwan university Cairo SWMC |

eC32 at 22°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 215mA/12V | TCD/Birr castle, Irland (instrument #3) |

eC33 at 22°C | plot at AGC=100pwm | sfdr | N/A | Sens | plot vs frequency plot vs frequency |
N/A |
221mA/12V | HB9SCT Modelboard/Demonstrationboard in open aluminum frame 2019-04-18 upgrade for bw=80KHz |

eC34 at 22°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 224mA/12V | Metsähovi Finland linear polarization LPDA |

eC35 at 22°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 220mA/12V | Metsähovi Finland linear polarization LPDA |

eC36 at 24.5°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 238mA/12V | TCD/Birr castle, Irland low frequency 20-90 MHz |

eC37 at 24.5°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 236mA/12V | Was in Anchorage, Alaska USA (on loan), then HRS-Box for Astrowoche Diavolezza and then observing 21cm radiation at suntower STS Zurich observatory. Firmware update 20190729 to V1.8 Replacement for broken eC41 to RCAG in Ulaan Baatar, Mongolia |

eC38 at 24.5°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 233mA/12V | TSAO, Almaty, Kazakhstan in Tian Shan mountains |

eC39 at 21°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 235mA/12V | UoM Mauritius. (U) |

eC40 at 24.5°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 231mA/12V | UoM Mauritius, (V) |

eC41 at 24.5°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 231mA/12V 229mA/12V |
RCAG Mongolia (2nd instrument). Repair 09.08.2017: MAX-232 and power-switch replaced. 2019-08-14 killed by lightning strike. Instrument will soon be replaced by a new one (eC37) |

eC42 at 24.5°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 221mA/12V | ANGKASA, National Space Centre, Malaysia |

eC43 at 21°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 234mA/12V | ANGKASA, National Space Centre, Malaysia |

eC44 at 24.5°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 240mA/12V | Crimean Astrophysical Observatory, Simpheropol, Ukraine |

eC45 at 21°C | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 607.19MHz | 232mA/12V | Badary Russian Federation (replacement for eC03) |

2nd series of Callisto spectrometer based on Philips CD1316LS/IV-3 which is compatible with CD1316L IHP-3. Firmware on ATmega16 is no more compatible to CD1316L/IV (HW manufactured by Valentina Waller and Jonas Voit), 29.06.2011 - 18.08.2011 | ||||||||

eC46 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | N/A | 249mA/12V | D. Shetti, Kasturbai Walchand College, Sangli, Maharashtra, India-416416 |

eC47 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | N/A | 253mA/12V | Pune, India |

eC48 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | N/A | 251mA/12V | Katubedda, Moratuwa, Sri Lanka |

eC49 | plot at AGC=100pwm | sfdr | NF | Sens | plot vs frequency | N/A | 251mA/12V | University of Nairobi, Kenya |

eC50 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | N/A | 248mA/12V | SCO, Hurbanovo, Slovakia |

eC51 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | N/A | N/A | DL6EBS -> 2017 -> OE3FLB, Fritz L. Austria |

eC52 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | N/A | 253mA/12V | INAF, Trieste, Italy (instruments #1) |

eC53 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | N/A | 254mA/12V | INAF, SRT rfi-monitoring Cagliari, Sardinia 2016 -> Turin (Fabrizio & Claudio) |

eC54 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | N/A | 261mA/12V | ANGKASA, National Space Centre, Malaysia |

eC55 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | N/A | 252mA/12V | ANGKASA, National Space Centre, Malaysia |

eC56 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | N/A | 259mA/12V | HB9SCT, Freienbach Switzerland, CD1316/IHP-3, 80 KHz, HAM-Version/Funkamateur |

3rd series of Callisto spectrometer based on Philips CD1316LS/IV-3 New LO-quartz 25.43 MHz, new firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Moritz St öckli), 15.06.2012 - 26.06.2012 | ||||||||

eC57 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 220mA/12V | University of Alcalá, Spain (instrument #1 for Peralejos) |

eC58 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 233mA/12V | INAF, Trieste, Italy (instrument #2) |

eC59 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 241mA/12V | Indonesian Institute of Aeronautics and Space, Division of Solar and Space Science in BANDUNG |

eC60 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 230mA/12V | INAF, Trieste, Ital (instrument #3) |

eC61 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 240mA/12V | University of Glasgow, Scotland UK |

eC62 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plotvs integration time at 610.0 MHz | 239mA/12V | Rajamangala University of Technology Krungthep, Bangkok, Thailand |

eC63 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 240mA/12V | TCD/Birr castle, Irland. Instrument #5 |

eC64 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 229mA/12V | TCD/Birr castle, Irland. Instrument #4 |

eC65 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 232mA/12V | Rajamangala University of Technology Krungthep, Bangkok, Thailand |

eC66 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 230mA/12V | Roztoky, Slovakia |

4th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Fabian Hofmann), 10.12.2012 | ||||||||

eC67 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 240mA/12V | Science College, Nashik, India |

eC68 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 245mA/12V | University of Alcalá, Spain (instrument #2 for Alcalá) |

eC69 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 240mA/12V | Bosscha Observatory, Institut Teknologi Bandung, Jl. Ganesha 10, Bandung 40132, Indonesia |

eC70 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 245mA/12V | University of Murcia, Sigüenza, Spain (instruments #3) |

eC71 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 250mA/12V | P. Keller Maturaarbeit mit LWA -> Kantonsschule Wattwil/SG -> Heiterswil |

5th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Manuela Wipf), 16.05.2013 | ||||||||

eC72 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 259mA/12V | Space Science Center, Universiti Kebangsaan Malaysia + LNA10 |

eC73 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 254mA/12V | Space Science Center, Universiti Kebangsaan Malaysia + LNA12 |

eC74 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 292mA/12V | University of Rwanda in Kigali, together with LNA13 + e-LNA08 |

eC75 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | 299mA/12V | SANSA Space Science, Hermanus, South Africa |

6th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Alexander Zvyagin), 08.05.2014 | ||||||||

eC76 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 60.5 MHz | 237mA/12V | Istituto Ricerche Solari Locarno (IRSOL) with LPDA, no LNA due to rfi. http://www.irsol.ch/ |

eC77 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 60.5 MHz | 236mA/12V | Institute of Ionosphere, Almaty Kazakhstan, 2nd Callisto + LNA15 |

eC78 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 60.5 MHz | 232mA/12V | UeNo & Nozawa, Hida Observatory, Kyoto University, Japan. Now Ibaraki University |

eC79 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 60.5 MHz | 230mA/12V | Ritma Harta Malaysia + LNA16 + LNA17 |

7th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Thomas Lanz), 14.05.2014 | ||||||||

eC80 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 60.5 MHz | ~230mA/12V | Ritma Harta Malaysia + LNA18 |

eC81 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 60.5 MHz | ~230mA/12V | Observatorio Astronómico Los Molinos (OALM), Montevideo, Uruguay |

eC82 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 60.5 MHz | ~230mA/12V | Blair Athol, Adelaide Australia |

8th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Mona Hänni), 02.10.2014 | ||||||||

eC83 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~226mA/12V | Bulgarian National Astronomical Observatory Rozhen + LNA19 |

eC84 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~231mA/12V | Space Science Center, Universiti Kebangsaan Malaysia + LNA20 |

eC85 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~229mA/12V | P. Hirt Muhen to play with LPDA |

eC86 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~247mA/12V | Delivered to Universidad Politecnica de Cartagena, Spain |

9th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Thomas Keller), 15.10.2015 | ||||||||

eC87 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~233mA/12V | Delivered to Universidad Politecnica de Cartagena, Spain |

eC88 small band width | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~235mA/12V | Kangerlussuaq, Kellyville Greenland (85KHz bandwidth). Up-converter LO=350 MHz. |

eC89 small band width | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~230mA/12V | Kangerlussuaq, Kellyville Greenland (85KHz bandwidth). Up-converter LO=350 MHz. |

eC90 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~234mA/12V | rfi -monitoring SKA HQ Cambridge by H. Smith |

10th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Manuela Wipf), 23.06.2016 | ||||||||

eC91 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~232mA/12V ~233mA/12V | R. Jain, Kadi Sarva Vishwavidyalaya, Gandhinagar, Ahmedabad, India 2017-07-04 In total 4 chips replaced after a lightning stroke. System fully tested. |

eC92 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | N/A | ~231mA/12V | Modul used for rfi-monitoring system in Brazil (BINGO-project) |

11th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Reto Schlatter), 09.02.2016 | ||||||||

eC93 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~256mA/12V | Lustbühel, University of Graz, Austria |

eC94 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~237mA/12V | Lustbühel, University of Graz, Austria |

eC95 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~248mA/12V | Station WARKWRTH, Auckland University of Tech., Auckland 1010, NEW ZEALAND |

eC96 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~236mA/12V | Metsähovi Metsähovi Radio Observatory, Kylmälä, Finland, LWA polarization x |

12th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Nicola Schneider), 25.01.1017 | ||||||||

cC97 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~248mA/12V | Metsähovi Metsähovi Radio Observatory, Kylmälä, Finland, LWA polarization y |

eC98 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~254mA/12V | DG8OV, HiGA Hildesheim with 10.7 MHz wide band IF-output |

13th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Bernard Leung), 29.03.2017 | ||||||||

eC99 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~245mA/12V | Telecom University, Indonesia |

eC100 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~242mA/12V | IIA, Kodaikanal, India |

eC101 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~246mA/12V | IIA, Kodaikanal, India |

14th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Marvin Wälti; reflow oven used for 1st time), 24.10.2017 | ||||||||

eC102 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~260mA/12V | P. Hirt Muhen (LWA + 90°-Hybrid + up-converter, LO=174 MHz) LHCP |

eC103 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~251mA/12V | P. Hirt Muhen (LWA + 90°-Hybrid + up-converter, LO=174 MHz) RHCP |

eC104 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~255mA/12V | Bhuwan J., Udaipur/India + e-LNA04 |

eC105 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~251mA/12V | Suman G., Pokhara/Nepal + e-LNA06. LNA replaced by SPF5189 18.02.2019 |

eC106 wideband 2.4 MHz | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~230mA/12V | Norwegian Space Centre + converter + frontend; 1-1.6 GHz->100-700 MHz |

eC107 wideband 2.4 MHz | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~236mA/12V | Norwegian Space Centre + converter + frontend; 1-1.6 GHz->100-700 MHz |

15th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Christian Monstein), KW11-12, 2018 | ||||||||

eC108 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~254mA/12V | Asnor UKM Malaysia + e-LNA07 |

eC109 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~257mA/12V | Metsähovi Metsähovi Radio Observatory, Kylmälä, Finland for testing and debugging |

eC110 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~257mA/12V | Arecibo, Puerto Rico + up-converter (LO=750 MHz), Delivery: 2019-03-11 |

eC111 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~252mA/12V | Arecibo, Puerto Rico + up-converter (LO=750 MHz), Delivery: 2019-03-11 |

16th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Christian Monstein), KW38, 2018 | ||||||||

eC112 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~256mA/12V | Sri Lanka ACCMT-SRI/SRI-Lanka, Delivery: 2019-02-29 |

eC113 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~250mA/12V | Sri Lanka ACCMT-SRI/SRI-Lanka, Delivery: 2019-02-29 |

eC114 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~245mA/12V | Astronomical Institute of the Romanian Academy, 2021-05-21 |

eC115 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~246mA/12V | Metsähovi Metsähovi Radio Observatory, Kylmälä, Finland for testing and debugging |

eC116 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~246mA/12V | Rajamangala University of Technology Krungthep, Bangkok, Thailand |

17th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Christian Monstein), KW04, 2019 | ||||||||

eC117 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~262mA/12V | Rajamangala University of Technology Krungthep, Bangkok, Thailand |

eC118 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~253mA/12V | Indian Institute of Astronomy IIA, Bangalore, India. 2020-01-17 |

eC119 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~253mA/12V | Satoshi Nozawa Earth Science of Ibaraki University Bunkyo 2-1-1, Mito, Ibaraki, Japan, 2020-02-09 |

eC120 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~256mA/12V | Storage @ HB9SCT |

18th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Christian Monstein), KW04, 2020 | ||||||||

eC121 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~262mA/12V | Strauss, North-West University, SA incl. e-LNA21, DiSEqC-module. |

eC122 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~245mA/12V | Strauss, North-West University, SA |

eC123 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~249mA/12V | Strauss, North-West University, SA |

eC124 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~250mA/12V | 2020-07-01 DLR Neustrelitz, Germany. |

19th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Christian Monstein), KW24, 2020 | ||||||||

eC125 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~277mA/12V | Alexandria, Egypt 2020-12-06. Heterodyne-up-converter with LO=125 MHz. |

eC126 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~262mA/12V | Alexandria, Egypt 2020-12-06. Heterodyne-up-converter with LO=125 MHz. |

eC127 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~269mA/12V | Bleien RT, ETH Zurich, 2021-06-15. |

eC128 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~248mA/12V | Bleien RT, ETH Zurich, 2021-06-15. |

eC129 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~274mA/12V | Reserved ETH Zurich. |

20th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Christian Monstein), KW51, 2020 - KW03, 2021 | ||||||||

eC130 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~261mA/12V | Delivery to IIA, 2021-04-12. Heterodyne up-converter CON018, LO=125 MHz. |

eC131 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~253mA/12V | Delivery to IIA, 2021-04-12. Heterodyne up-converter CON019, LO=125 MHz. |

eC132 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~252mA/12V | Delivery to IIA, 2021-04-12. Heterodyne up-converter CON020, LO=125 MHz. |

eC133 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~256mA/12V | Delivery to IIA, 2021-04-12. |

21th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Christian Monstein), KW23, 2021 | ||||||||

eC134 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~240mA/12V | Delivered to Baharuddin Bin Jamal Mohamed, Malaysia together with LNA14, 2021-08-16. |

eC135 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~240mA/12V | Delivered to Urs G. Kantonsschule Wattwil. 2021-08-31. |

eC136 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~240mA/12V | Delivered to Agensi Angkasa Selangor, Banting, Malaysia + LNA22 + LNA2000/3 2021-09-03. |

eC137 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~240mA/12V | Bosscha Observatory, ITB with dual channel heterodyne up-converter CON024, LO=175 MHz. 2021-09-17. |

22th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Christian Monstein), KW23-KW30, 2021 | ||||||||

eC138 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~240mA/12V | Bosscha Observatory, ITB with dual channel heterodyne up-converter CON024, LO=175 MHz. 2021-09-17. |

eC139 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~240mA/12V | DLR Neustrelitz, Germany with dual polarization heterodyne up-converter CON025. |

eC140 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~250mA/12V | DLR Neustrelitz, Germany with dual polarization heterodyne up-converter CON025. |

eC141 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~250mA/12V | Storage @ HB9SCT |

23th series of Callisto spectrometer based on Philips CD1316LS/IV-3. LO-quartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Christian Monstein), KW37-KW40, 2021 | ||||||||

eC142 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~250mA/12V | Storage @ HB9SCT |

eC143 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~240mA/12V | Storage @ HB9SCT |

eC144 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~240mA/12V | Storage @ HB9SCT |

eC145 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~250mA/12V | Storage @ HB9SCT |

eC146 | plot at AGC=80pwm | sfdr | NF | Sens | plot vs frequency | plot vs integration time at 610.0 MHz | ~240mA/12V | Storage @ HB9SCT |

Legend:

All qualification tests are performed under similar conditions with an automatic test-system, based on IEEE-488 instrumentation.
Power supply, clock generator, noise test system, rf-signal generator, notebook, DUT