Callisto Qualification with test results from the rflaboratory
Serial number temperature +/ 1°C 
Detector slope 
SFDR 
Noise Figure 
Sensitivity limit 
Band pass filter 
Allan time, Allan variance 
Supply current 
Location, user, comments and links 
1st series of Callisto spectrometer based on Philips CD1316/IV. (HW manufactured by A. James, R. Moser, P. Behm, I. Zamora, S. Zihlmann, M. Nyirö, H. Temperli, M. Battaglia, A. Vydrzel, T. Moser and T. Kittelmann), 10.07.2009  31.05.2010  
eC00 (breadboard) 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
225mA/12V 
used for spare parts ... 
eC01 (prototype) at 24°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs time 607.19MHz
(channel 130) full report 
222mA/12V 
HPTB109 
eC02 at 23.5°C 
plot vs AGCvoltage 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs time 458.69MHz
(channel 100) full report 
223mA/12V 

eC03 at 22.5°C 
plot vs AGCvoltage 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs time 607.19MHz
(channel 130) full report 
229mA/12V 
University of Addis Ababa (AAU), Ethiopia 
eC04 at 22°C 
plot at PWM=120 
sfdr 
NF 
Sens 
plot vs frequency 
N/A 
226mA/12V 
HB9SCT, Freienbach 
eC05 at 22°C 
plot at PWM=120 
sfdr 
NF 
Sens 
plot vs frequency 
N/A 
230mA/12V 
Bleien RSG, ETH Zürich (7m LHCP) 
eC06 at 22°C 
plot at PWM=120 
sfdr 
NF 
Sens 
plot vs frequency 
N/A 
242mA/12V 

eC07 at 22°C 
plot at diff. PWM 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs time 607.19MHz 
226mA/12V 
KASI Daejeon
South 
eC08 at 21°C 
plot at diff. PWM 
sfdr 
NF 
1) 
plot vs frequency 
N/A 
220mA/12V 
KASI 
eC09 at 21°C 
plot at diff. PWM 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs time 800MHz 
216mA/12V 
Royal Observatory of 
eC10 at 21°C 
plot at diff PWM 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs time 800MHz 
218mA/12V 
Ooty 
eC11 at 21°C 
plot at diff PWM 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs time 800MHz 
217mA/12V 
Ooty 
eC12 at 21°C 
plot at diff PWM 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs time 800MHz 
215mA/12V 
RSG Bleien SBand 5m, derzeit als Ersatz RHCP 7m 
eC13 at 21°C 
plot at diff PWM 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs time 800MHz 
214mA/12V 
OSRA, Ondrejov, Czech Republic 
eC14 at 21°C 
plot at diff PWM 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs time 800MHz 
215mA/12V 
Royal Observatory of 
eC15 at 21°C eC15a at 20°C 
plot at diff PWM plot at PWM=80 
sfdr sfdr 
NF NF 
Sens Sens 
plot vs frequency plot vs frequency 
plot vs time 800MHz nil 
213mA/12V 236mA/12V 
UoM Repair and upgrade 20170426 (all IC, Quartz 25.43 MHz, R3, R4) 
eC16 at 21°C 
plot at diff PWM 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs time 800MHz 
213mA/12V 
Gauribidanur

eC17 at 21°C 
plot at diff PWM 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs time 800MHz 
214mA/12V 213mA/12V 
RCAG Test 09.08.2017, everything o.k. 
eC18 at 21°C 
plot at diff PWM 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs time 800MHz 
212mA/12V 
Travel set in aluminum case HIT G13.2 
eC19 at 19°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs integration time at
607.19MHz 
233mA/12V 
TCD 
eC20 at 20°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs integration time at
607.19MHz 
219mA/12V 
INPE São José dos 
eC21 at 18°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs integration time at
607.19MHz 
222mA/12V 
Bleien RSG, ETH Zürich Bicone 1080 MHz 
eC22 at 20°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs integration time at
607.19MHz 
219mA/12V 
HPP Q12, Praktikum + Astrowoche 2) 
eC23 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
Kit for 
eC24 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
N/A 
Kit for 
eC25 at 23°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs integration time at
607.19MHz 
222mA/12V 
RSG Ersatzgerät für LBand 5m 
eC26 at 23°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs integration time at
607.19MHz 
222mA/12V 
HPTB109 ex. Modelboard/Demonstrationboard in open aluminum frame 
eC27 at 23°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs integration time at
607.19MHz 
220mA/12V 
ERAC 
eC28 at 23°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs integration time at
607.19MHz 
222mA/12V 
INPE São José dos 
eC29 at 23°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot vs frequency 
plot vs integration time at
607.19MHz 
220mA/12V 
Jim Sky, 
eC30 at 22°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot vs frequency 
plot
vs integration time at 607.19MHz 
215mA/12V 
DL2RD Berlin > ANGKASA Malaysia 
eC31 at 22°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot
vs frequency 
plot vs integration time at
607.19MHz 
215mA/12V 
Helwan university Cairo SWMC 
eC32 at 22°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot
vs frequency 
plot vs integration time at
607.19MHz 
215mA/12V 
TCD/Birr castle, Irland (instrument #3) 
eC33 at 22°C 
plot at AGC=100pwm 
sfdr 
N/A 
Sens 
plot vs frequency 
N/A 
221mA/12V 
HPTB109 ex. Modelboard/Demonstrationboard in open aluminum frame 
eC34 at 22°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot
vs frequency 
plot vs integration time at
607.19MHz 
224mA/12V 
Metsähovi Finland linear polarization LPDA 
eC35 at 22°C 
plot at AGC=100pwm 
sfdr 
NF 
Sens 
plot
vs frequency 
plot vs integration time at
607.19MHz 
220mA/12V 
Metsähovi Finland linear polarization LPDA 
eC36 at 24.5°C 
plot at AGC=100pwm 






TCD/Birr castle, Irland low frequency 2090 MHz 
eC37 at 24.5°C  plot at AGC=100pwm 






Was in Anchorage, Alaska USA (on loan), then HRSBox for Astrowoche Diavolezza and now observing 21cm radiation at suntower STS Zurich observatory 
eC38 at 24.5°C 
plot at AGC=100pwm 






TSAO, Almaty, Kazakhstan in Tian Shan mountains 
eC39 at 21°C 
plot at AGC=100pwm 






UoM 
eC40 at 24.5°C 
plot at AGC=100pwm 






UoM 
eC41 at 24.5°C 
plot at AGC=100pwm 





229mA/12V 
RCAG Mongolia (2nd instrument) Repair 09.08.2017: MAX232 and powerswitch replaced, rest o.k. 
eC42 at 24.5°C 
plot at AGC=100pwm 






ANGKASA, National Space Centre, Malaysia 
eC43 at 21°C 
plot at AGC=100pwm 






ANGKASA, National Space Centre, Malaysia 
eC44 at 24.5°C 
plot at AGC=100pwm 






Crimean Astrophysical Observatory, Simpheropol, Ukraine 
eC45 at 21°C 
plot at AGC=100pwm 






Badary
Russian Federation (replacement for eC03) 
2nd series of Callisto spectrometer based on Philips CD1316LS/IV3 which is
compatible with CD1316L/IHP3. Firmware on ATmega16 is no more
compatible to CD1316L/IV (HW manufactured by Valentina Waller and Jonas Voit), 29.06.2011  18.08.2011  
eC46  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  N/A  249mA/12V  D. Shetti, Kasturbai Walchand College, Sangli, Maharashtra, India416416 
eC47  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  N/A  253mA/12V  Pune, India 
eC48  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  N/A  251mA/12V  Katubedda, Moratuwa, Sri Lanka 
eC49  plot at AGC=100pwm  sfdr  NF  Sens  plot vs frequency  N/A  251mA/12V  University of Nairobi, Kenya 
eC50  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  N/A  248mA/12V  SCO, Hurbanovo, Slovakia 
eC51  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  N/A  N/A  DL6EBS http://www.radioastronomie.de 2017 > OE3FLB, Fritz L. Austria 
eC52  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  N/A  253mA/12V  INAF, Trieste, Italy (instruments #1) 
eC53  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  N/A  254mA/12V  INAF, SRT rfimonitoring Cagliari, Sardinia 2016 > Turin (Fabrizio & Claudio) 
eC54  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  N/A  261mA/12V  ANGKASA, National Space Centre, Malaysia 
eC55  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  N/A  252mA/12V  ANGKASA, National Space Centre, Malaysia 
eC56  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  N/A  259mA/12V 
HB9SCT, Freienbach 
3rd
series of Callisto spectrometer based on Philips CD1316LS/IV3 New
LOquartz 25.43 MHz, new firmware V1.8. For PC software go for version
V116 or higher (HW manufactured by Moritz Stöckli), 15.06.2012  26.06.2012  
eC57  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  220mA/12V  University of Alcalá, Spain (instrument #1 for Peralejos) 
eC58  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  233mA/12V  INAF, Trieste, Italy (instrument #2) 
eC59  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  241mA/12V  Indonesian Institute of Aeronautics and Space, Division of Solar and Space Science in BANDUNG 
eC60  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  230mA/12V  INAF, Trieste, Italy (instrument #3) 
eC61  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  240mA/12V  University of Glasgow, Scotland UK 
eC62  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  239mA/12V  Part of my current international travelset 
eC63  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  240mA/12V  TCD/Birr castle, Irland. Instrument #5 
eC64  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  229mA/12V  TCD/Birr castle, Irland. Instrument #4 
eC65  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  232mA/12V  HIT office (international travel set) 
eC66  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  230mA/12V  Roztoky, Slovakia 
4th series of Callisto spectrometer based on Philips CD1316LS/IV3. LOquartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Fabian Hofmann), 10.12.2012  
eC67  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  240mA/12V  Science College, Nashik, India 
eC68  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  245mA/12V  University of Alcalá, Spain (instrument #2 for Alcalá) 
eC69  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  240mA/12V  Bosscha Observatory,Institut Teknologi Bandung, Jl. Ganesha 10, Bandung 40132, Indonesia 
eC70  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  245mA/12V  University of Murcia, Sigüenza, Spain (instruments #3) 
eC71  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  250mA/12V  P. Keller Maturaarbeit mit LWA > Kantonsschule Wattwil/SG 
5th series of Callisto spectrometer based on Philips CD1316LS/IV3. LOquartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Manuela Wipf), 16.05.2013  
eC72  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  259mA/12V  Space Science Center, Universiti Kebangsaan Malaysia + LNA10 
eC73  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  254mA/12V  Space Science Center, Universiti Kebangsaan Malaysia + LNA12 
eC74  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  292mA/12V  University of Rwanda in Kigali, together with LNA13 
eC75  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  299mA/12V  SANSA Space Science, Hermanus, South Africa 
6th series of Callisto spectrometer based on Philips CD1316LS/IV3. LOquartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Alexander Zvyagin), 08.05.2014  
eC76  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 60.5 MHz  237mA/12V  HPTB109 used to perform tests in the laboratory 
eC77  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 60.5 MHz  236mA/12V  Institute of Ionosphere, Almaty Kazakhstan, 2nd Callisto + LNA15 
eC78  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 60.5 MHz  232mA/12V  UeNo & Nozawa, Hida Observatory, Kyoto University, Japan. Now Ibaraki University 
eC79  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 60.5 MHz  230mA/12V  Ritma Harta Malaysia + LNA16 + LNA17 
7th series of Callisto spectrometer based on Philips CD1316LS/IV3. LOquartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Thomas Lanz), 14.05.2014  
eC80  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 60.5 MHz  ~230mA/12V  Ritma Harta Malaysia + LNA18 
eC81  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 60.5 MHz  ~230mA/12V  Observatorio Astronómico Los Molinos (OALM), Montevideo, Uruguay 
eC82  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 60.5 MHz  ~230mA/12V  Blair Athol, Adelaide Australia 
8th series of Callisto spectrometer based on Philips CD1316LS/IV3. LOquartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Mona Hänni), 02.10.2014  
eC83  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~226mA/12V  Bulgarian National Astronomical Observatory Rozhen + LNA19 
eC84  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~231mA/12V  Space Science Center, Universiti Kebangsaan Malaysia + LNA20 
eC85  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~229mA/12V  Observatory Heiterswil: Interferometer testing by P. Keller > QTH de HB9SCT 
eC86  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~247mA/12V  Delivered to Universidad Politecnica de Cartagena, Spain 
9th series of Callisto spectrometer based on Philips CD1316LS/IV3. LOquartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Thomas Keller), 15.10.2015  
eC87  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~233mA/12V  Delivered to Universidad Politecnica de Cartagena, Spain 
eC88 small band width  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~235mA/12V  Kangarlussuaq, Kellyville Greenland (85KHz bandwidth) 
eC89 small band width  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~230mA/12V  Kangarlussuaq, Kellyville Greenland (85KHz bandwidth) 
eC90  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~234mA/12V  rfi monitoring SKA HQ Cambridge by H. Smith 
10th series of Callisto spectrometer based on Philips CD1316LS/IV3. LOquartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Manuela Wipf), 23.06.2016  
eC91  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~232mA/12V ~233mA/12V  R. Jain, Kadi Sarva Vishwavidyalaya, Gandhinagar, Ahmedabad, India 20170704 In total 4 chips replaced after a lightning stroke. System fully tested. 
eC92  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  N/A  ~231mA/12V  HPTB109 modul used for rfimonitoring system in Uruguay or Brazil (BINGO) 
11th series of Callisto spectrometer based on Philips CD1316LS/IV3. LOquartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Reto Schlatter), 09.02.2016  
eC93  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~256mA/12V  Lustbühel, University of Graz, Austria 
eC94  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~237mA/12V  Lustbühel, University of Graz, Austria 
eC95  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~248mA/12V  Station WARKWRTH, Auckland University of Tech., Auckland 1010, NEW ZEALAND 
eC96  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~236mA/12V  Metsähovi Radio Observatory, Kylmälä, Finland, LWA polarization x 
12th series of Callisto spectrometer based on Philips CD1316LS/IV3. LOquartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Nicola Schneider), 25.01.1017  
cC97  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~248mA/12V  Metsähovi Radio Observatory, Kylmälä, Finland, LWA polarization y 
eC98  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~254mA/12V  DG8OV, HiGA Hildesheim with 10.7 MHz wide band IFoutput 
13th series of Callisto spectrometer based on Philips CD1316LS/IV3. LOquartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Bernard Leung), 29.03.2017  
eC99  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~245mA/12V  Telcom University, Indonesia 
eC100  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~242mA/12V  IIA, Bangalore, India 
eC101  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~246mA/12V  IIA, Bangalore, India 
14th series of Callisto spectrometer based on Philips CD1316LS/IV3. LOquartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Marvin Wälti; reflow oven used for 1st time), 24.10.2017  
eC102  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~260mA/12V  P. Hirt Muhen (LWA + 90°Hybrid + upconverter, LO=174 MHz) LHCP 
eC103  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~251mA/12V  P. Hirt Muhen (LWA + 90°Hybrid + upconverter, LO=174 MHz) RHCP 
eC104  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~255mA/12V  Bhuwan J., Udaipur/India + eLNA04 
eC105  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~251mA/12V  Suman G., Pokhara/Nepal + eLNA06 
eC106 wideband 2.4 MHz  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~230mA/12V  Norwegian Space Centre + converter + frontend; 11.6 GHz>100700 MHz 
eC107 wideband 2.4 MHz  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~236mA/12V  Norwegian Space Centre + converter + frontend; 11.6 GHz>100700 MHz 
15th series of Callisto spectrometer based on Philips CD1316LS/IV3. LOquartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Christian Monstein), KW1112, 2018  
eC108  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~254mA/12V  HPTB109 reservation for Malaysia + eLNA07 
eC109  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~257mA/12V  HPTB109 
eC110  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~257mA/12V  HPTB109 
eC111  plot at AGC=80pwm  sfdr  NF  Sens  plot vs frequency  plot vs integration time at 610.0 MHz  ~252mA/12V  HPTB109 
16th series of Callisto spectrometer based on Philips CD1316LS/IV3. LOquartz crystal 25.43 MHz, Firmware V1.8. For PC software go for version V116 or higher (HW manufactured by Christian Monstein), KW38, 2018  
eC112  plot at AGC=80pwm  sfdr  tbd  Sens  plot vs frequency  tbd  ~256mA/12V  In production HPTB109 
eC113  plot at AGC=80pwm  sfdr  tbd  Sens  plot vs frequency  tbd  ~250mA/12V  In production HPTB109 
eC114  plot at AGC=80pwm  sfdr  tbd  Sens  plot vs frequency  tbd  ~245mA/12V  In production HPTB109 
eC115  plot at AGC=80pwm  sfdr  tbd  Sens  plot vs frequency  tbd  ~246mA/12V  In production HPTB109 
eC116  plot at AGC=80pwm  sfdr  tbd  Sens  plot vs frequency  tbd  ~246mA/12V  In production HPTB109 
Legend:
1) Difficult measurement lead to uncertain results, software was not
appropriate at that time.
2) Additional capacitors can be switched in for higher integration time
(10ms .... 3sec). Processor now 10bit ADC based on ATmega16, V1.6
3) if (during download) the file should be saved as *.zip then just rename it to
*.jar Prior to that you need to install Java Runtime system.
Version: 20180928
Responsible: Christian Monstein, cmonstei(at)retired.ethz.ch